SOP on cleaning, maintenance and calibration of Cooling cabinet
- To lay down a procedure for Cleaning, maintenance and calibration of Cooling cabinet.
- This SOP is applicable for the cleaning, maintenance and calibration of cooling cabinet
- QC Chemist or above
- Electrician or above
- Head – Quality Control
- Head – Engineering
- Only trained and skilled persons shall maintain the cleaning of Cooling cabinet.
- Cleaning of cooling cabinet shall be done once in a day.
- PT-100 sensor shall be cleaned with polish paper in every four month.
- Condensor shall be Clean by blowing air through it after every four months.
- Cleaning record of the cooling cabinet shall be maintained as per Annexure No.A01.
- Any problem in stability chamber shall be intimated to Engineering department and fill the maintenance work permit as per Annexure No. A02.
- Chamber fan and condensor fan shall be checked in four months. If found loose, tighten it.
- Working of safety cut off shall be checked in four month by giving external heat to the dry bulb sensor.
- Maintenance shall be done if any problem appears in the chamber.
- Preventive maintenance of stability chamber shall be done by the service engineer in four months and record shall be maintained as per Annexure NoA06.
- If any abnormality observed by service engineer then action for rectification shall be done immediately.
- Check the incoming electric supply of 230 V AC with Neutral and proper earthing.
- Maintenance record shall be maintain as per Annexure No. A02
|COOLING EFFECT NOT ACHIEVED WITH REGULAR COOLING SYSTEM|
|MCB may have trip.||Check the MCB5 (604 & 605), if found trip, switch it ON.|
|DC output not available.||Check the DC output from the temperature controller.|
|SSR not working.||The DC supply comes from the temperature controller terminal 13 (351) and 14 (352) to SSR terminal 3(+) & 4(-). The output of the SSR is provided with a phase connection coming from the Time delay and the other output goes to the compressor. If the controller is giving DC volt to the SSR input and the SSR output terminal is (606) not showing the phase (230 Volts) that means the SSR is not working, this has to be replaced.|
|Overload or the capacitor of the compressor not working.||Check the compressor for overload and starting and running capacitor for the supply to the compressor. If found faulty, replace it.|
|Time delay faulty|
|The supply from compressor switch (S/W5) is given to the time delay. The output of the time delay is given to the compressor through relay. Check the supply at 604. If the supply is not coming then NO point is not working. Hence time delay has to be replaced.|
|Condenser Motor not working||Check the motor winding.|
CHAMBER MOTOR NOT WORKING
|MCB may have trip.||Check the MCB1 (401 & 402). If found trip, Switch it ON.|
|Starting capacitor not working.||Check the starting capacitor, If found faulty, replace it.|
|Motor burned out.||Check the Motor winding, if found burned out or found any problem replace it.|
|CONTROLLER NOT WORKING|
|MCB may have blown off.||Check the MCB2(312 & 313).If found blown replace it.|
|Display card or inside component loose||If the display card or any of the inside components is loose, the controller will not show any display or will work erratic. Replace the controller.|
|CVT not working||Check the CVT out put Voltage, it should be 230 VAC. If it is not obtained then replace the CVT.|
|CONTROLLER SHOWING OPEN|
|Sensor wire broken.||Check the sensor wire. If found broken, replace the sensor.|
|Sensor wire having loose connections.||Check the sensor wire connections on the controller. If found loose, tighten it.|
- Calibration of the stability chamber shall be done by outside party once in a year.
- Calibration shall be reported as per Annexure No. A02
List of Annexure / Format :
|1|| Cooling cabinet Cleaning|
|2|| Cooling cabinet Maintenance|
|3||Cooling Cabinet calibration|
- Cooling cabinet Manual.
Reason for Revision :
- Not applicable due to first version of SOP
- QC : Quality control
- SOP : Standard Operating Procedure
Cooling Cabinet cleaning Record
Cooling Cabinet Id. No.:
|Cleaning done by||Checked by|
Cooling Cabinet maintenance record
Cooling Cabinet Id. No. : Date :
|Cooling Cabinet Model No.||Capacity|
Problem Identified :
Action for rectification :
Is calibration required after maintenance :
Remark ( if any ) :
|Service Engineer ( Sign / Date )||Checked By ( Sign / Date )||Approved By ( Sign / Date )|
Calibration record of Cooling Cabinet
Cooling Cabinet Id. No.:
Calibration Done On : Next Calibration due On :
|Set Temperature||Actual Temperature||Calibrator Temperature|
The above equipment has been found / not found within the permissible limits for temperature ±2°C .
Calibrator Details :
|Calibrator No.||Detail of Calibrator||Calibration Details||Calibration due date|
Remark if Any :
Instrument Calibration comply / does not comply.
|Reported By / Date||Checked By / Date||Approved By / Date|